PZ Laser Diode Stack With Sapphire. Spare parts for cosmetic lasers. Online advice on the repair of 808 nm diode lasers.
High reliability and stability of the laser bars
Number of shots: 30 millions
Compact designs,easy to integrate
Macro channel with simple water cooling and no strict requirement for water quality
Special shell size with light guide cone for PZ Laser handpiece
Center Wavelength 808nm
Wavelength Tolerance ±10nm
Working Mode QCW
Output Power Per Bar 100W
Number of Bars 10
Bar to Bar Spacing 2mm
Pulse Width 1~400ms
Duty Cycle 0.1~40%
Operating Current Iop 100A
Threshold Current Ith
Operating Voltage Vop 11V
Operating Temperature 20~30℃
Storage Temperature 0-55℃
Coolant Distilled Water Flow Rate 4~6L/min
Max Inlet Pressure 350~500kPa
Water Temperature(No Condensation) 23~27℃
Filtered Water Particles <20μm
The photographs show an example of a burnt diode stack of a PZ laser.
If you are interested in spare parts for lasers, then you can consult me to order a new diode stack. I will also conduct an online consultation on the replacement of this part.
It should be noted that various malfunctions of cosmetic lasers can be distinguished. Perhaps in your case the problem is in the electronic part. If you have any questions, you can contact me. I have extensive experience in the repair of diode lasers. I will also be able to connect you with the supplier of these parts to expedite the process.